**Home ⇒ Overview Courses ⇒ Sensor technology ⇒ Wheatstone Bridge ⇒ Wheatstone bridge with strain gauge **

### Measuring bridge with strain gauges

Table of Contents

Toggle**Strain gauges** were invented by **Arthur Claude Ruge** in 1938 in Massachusetts, USA. He recognized the linear relationship between the linear expansion of a conductor and its electrical resístance, and how this could be used to measure bending and compression.

The construction of an electrical strain gauge is simple: a wafer-thin conductor is placed on a carrier foil. Constantan is often used for the conductors because of its high temperature stability.

A fundamental parameter of the strain gage is the so-called **g**age **f**actor (GF), which describes its sensitivity to strain. The GF is defined as the ratio of the fractional change in electrical resístance ΔR / R to the fractional change in length, or strain ε:

**GF = ΔR / R | ε = ΔL / L**

**. ΔL / L**

**=> GF = ΔR / R or ΔR = GF * ε * R**

**. ε**

**Example:** The expansion of ε = 0.2% of a strain gauges with a GF of 2 results in an increase in resístance of 0.4%.

#### Quarter-, Half-, Full-bridge

Usually 1 or 2 or 4 strain gauges are installed in a Wheatstone bridge. The names quarter, half and full bridge result from this. The nominal resístance R0 of all strain gauges are usually the same. Typical values in the unloaded state are R0 = 120, 350, 700 or 1000 Ω.

The full bridge has the highest sensitivity (4 times the sensitivity compared to the quarter bridge). Since disturbances such as temperature drift have the same effect on all four strain gauges, this results in an effective compensation of these disturbance variables. When installing the strain gauges, make sure that two are stretched and two are compressed.

#### Work Order: Full bridge with strain gauges

- The strain gauges used in a full bridge have a resístance R
_{0}of 700 Ω, a maximum strain ε of 4 mm/m and a gage-factor of 2.1. What is the maximum resístance change ΔR of the strain gauges?

Calculate the maximum bridge voltage V_{BR}for this case. The supply voltage V of this measuring bridge is 10 V.

1) ΔR = GF * ε * R = 2,1 ¦ ε = 4 mm/m = 0,004 m / m = 0,004 = 0,4 %

. => ΔR = 2,1 * 0,004 * 700 Ω = 5,88 Ω

2) R_{1} = R_{4} = R_{0} - ΔR = 700 Ω - 5,88 Ω = 694,12 Ω

. R_{2} = R_{3} = R_{0} + ΔR = 700 Ω + 5,88 Ω = 705,88 Ω

V_{2} = R_{2} × V = 705,88 Ω × 10 V = 5,042 V

. R_{1} + R_{2} 1400 Ω

dito V_{4} = 4,958 V

=> V_{BR} = U2 - U4 = 0,084 V